In ADE7816, the no load current channels are showing high current readings. What might be the reason?
Please take a look at our technical support guidelines on the main page, https://ez.analog.com/community/energy-metering. We would need more information to understand what exactly the problem is and where it could be coming from.
If you have not performed a calibration yet, please have a read through the Calibration Application Note, this may solve your problem the fastest.
Retrieving data ...