We are going to use half bridge transducer with AD598.
Below is sensor spec. If anyone can review our schematic, please leave e-mail address.
The AD598 should not be used with a half bridge. It relies upon the sum of the secondary signals to be constant with stroke length. This is not possible if one channel is constant and one is moving. You can consider the AD698, which is intended for use with half bridge circuits. Please see figure 7 of the AD698 for additional details.
Retrieving data ...