I am using the ADL5802 to down convert two signals for phase comparison using the AD5302.
It appears the IF signals are 180 degrees apart when two in phase signals are put in.
Is this expected?
Judging from the Evaluation Board schematic, I'd say that we just have not been very rigorous in naming the IF output connectors: OP1+ and OP2+ on the IC should be in phase, but the naming of the IF2P & IF2N connectors have been reversed.
Retrieving data ...