I use the ADAS1000 as the ECG sampling frontend IC. But there always is sharp noise in the signal as follows. How should I eliminate it? I have enabled the RL driving and Shield driving.
So many thanks!
message duplicated, responses here.
thanks for message.
The screen shot show that the measurement is ok for some time and then something disturbs the signal.
Do you know what is happening at the time of this "noise"?
Is there motion during the measurement? Is there some interferer present at the electrode side during this time that might explain the noise?
Is your configuration based on the ADI evaluation board or your own hardware?
What is the signal source? patient simulator or person?
Please provide as much information as possible in order to help understand what might be going on.
Thanks in advance.
I also met the same problem, have you resolve this problem?
Retrieving data ...