I am having difficulties in doing a good bridge diagnostics. The indications in the UG does not match with the values obtained by AD conversion.
One of the scenario exemplified in table 15 page 27:
ADC changes by ΔV = +50 μA × R1. For example, ~100 mV for R1 = 2 kΩ.
The indication in the UG is missleading and wrong.
This is how I see the equation. Please correct me if I am wrong:
ΔV = V0 - V1
When no diag current enabled, and the bridge is in equilibrium, V0= V1 = Vdd/2
If we enable 50uA current source for ADC0, then V0 will become:
V0 = 50uA * R1 + (50uA + i1) * Rbridge;
i1 is the current in the higher left side of the bridge.
I have tested with 2.2 Kohm R1 and R2, and 2.2Kohm bridge. the delta is 165mV.
Can a UG errata be created to show exactly how this should be used?