[full disclosure] I am new to amplifiers and strain gauges.
I have an AD7730 eval board connected to a Vishay full-bridge strain gauge.
5V from power supply goes to DVDD, AVDD, Ref IN(+), and the top of the strain bridge.
GND goes to AGND, DGND, Ref IN(-), and the "bottom" of the strain bridge.
The other two taps of the full bridge go to AIN1(+) and ANIN1(-)
I am using the EVAL software Version 2.0.6 dated February 2002 (!!!!).
I cannot seem to set any of the program register parameters to show anything but FFFFFF in the "Read Display" screen of the EVAL software.
Any help, or am I way out in left field somewhere?
Thanks in advance,