the datasheet says:
±10 V range ±10 LSB
±5 V range ±90 LSB
for the TUE spec. The 10 LSB for the ±10 V range just look... wrong.
Should it be something like 100 LSB? Or 190 LSB?
The values on the datasheet are correct.
I need to confirm the values on the datasheet. Sorry that I cannot provide you with the answer right now, but I will get back to you in the next couple days.
Apologies for the delayed response.
The TUE is a combination of INL, Offset and gain error. The AD7609 is calibrated at +/-10V and result to very low gain and offset error at 25C therefore have a much better TUE .
so you can confirm the numbers in the datasheet?
OK. Thanks Jonathan.
Retrieving data ...