I'm looking for configuring an AD9106 for automatic calibration,
I provide a clock to the dac 10 mhz,
what value I have to write in the field "CAL_CLK_DIV" for I have 500KHZ?
how do I change the value in register 0x08 ?
Hi Roberto -
Please try a CAL_CLK_DIV setting of 0x7. In the lab we've found that the effect calibration has on gain accuracy is very small.
I have another question, I use datasheet REV.A, in the register summary is written that
SRAM_DATA (add0x6000 to 0x6FFF) are mapped : "15:12 reserved" and "11:0 sram_data",
but this is not correct, the right map is "15:4 sram_data" and "3:0" reserved.
Furthermore is data signed?
We're aware of this and in the process of updating the data sheet to rev B.
Retrieving data ...