I work on AD9106EBZ.
I would want to put an offset (DACx_DOFS) on one output without clipping output signal. Is it possible?
The platforme Labview is very badly made, an update is it available?
The to program the AD9106 I recommend editting and running the regval scripts that come with the AD9106 EVB software. The labview GUI is not robust enough for experimentation. The regvakl files can be found in C:\Program Files (x86)\Analog Devices\HSDAC\AD9106\AD9106\regval_files. You can run an editted regval file of your own by selecting the “default” option under Load Files(*.seq) and the “Search” option under Load Files(*.regval). You then select your regval file and press the “load register from sequence” button.
You can do DACxDOF without clipping as long as you leave yourself enough headroom.
Retrieving data ...