this image is from UG-061 eval board user guide meant for connecting with the mains. here, if VN is neutral voltage, why isn't P3 2 connected to neutral??
Sorry, we don't currently have a community on EngineerZone providing support for Analog Devices’ energy meter products. To get answers to your questions, I suggest contacting ADI support in your area. You can see the support contacts listed on our website at: www.analog.com/support.
I apologize for the inconvenience.
Retrieving data ...