I received a question about LTC2986 from our customer. He is checking LTC2986 by following setting.
When the customer sets YOKOGAWA 2553A to 0.0 deg C. The measurement result of LTC2986 is -0.5 deg C.
Then he sets 2553A to 100.0 deg C. The measurement…
I'm looking through the LTC2986 datasheet and I'm a little bit confused about the external configuration for the single ended and differential thermistors.
If I understand correctly, the device has ten channels, but can only be connected to three…
I have a couple of questons about the direct ADC mode of LTC2986. Please take care of this case.
- What is the maximum sampling speed of internal 24 bit ADCs in the direct ADC mode?
- How can I know conversion start and finish in the direct ADC mode…
About the way to speed up by using two LTC2986s, how can I get 12ch in 1s since this part's ODR is 164ms typ in the two conversion cycle mode?
I take back my question about the bootloader.
Hi @dariaa - apologies for the delayed response.
1) Leakage current will affect sensors and filters with high series resistance. My article goes into more details but you may see degraded accuracy with large series resistances at hot - keeping any filter…
Update:Just to be specific about what pins I’m talking about and how we’re using them I made this list:LTC2449: ~CS Our designs: Tied to GND / Return From datasheet: A LOW-to-HIGH transition on CS during the Data Output aborts the data transfer…