I used the ADT7310, but I don't know if the ADT7310 can be used with the ADT7310. If not, please recommend a temperature sensor for temperature measurement in a vacuum…
What kind of technology does the ADT7310 use? adt7310
We have 10 ADT7310 Temp.Sensors who works on 10 prototyps.
But we have following problems.
The ADT7310 works not in spezification of repeatability !!
All 10 Prototypes have the same problems on different temperatur !
Sensor 1-8 : (Ramp-up Temp…
I have an application requiring exceedingly stable temperatures ~0.001C would be great but I would settle for an order of magnitude would be acceptable. To measure temps, an ADT7310 had been chosen. I am a little unclear about the exact meaning…
May I ask you to send me an IBIS model for ADT7310?
Unfortunately we don't have an IBIS model for ADT7310.
Update: IBIS model for the ADT7310 is now available here: https://www.analog.com/en/design-center/simulation-models/ibis-models.html
I have following communication problem between STM32 Controller and ADT7310:
First time I will set the ADT7310 to continuous read mode with send command byte 0x54 to ADT7310.
But the ADT7310 switched to this mode only by multiply send 0x54 !!!
My sequence for read continuous read is:
first step init the ADT7310 with following commands:
1. Reset ADT7310 with 32 clocks on DIN
2. Set ADT7310 to 16 bit = send 1.Byte 0x08 (reg adress 01 / config reg.) send 2.Byte 0x80 to set res.to 16bit
Hi! Is the DOUT pin in high impedance state when CS pin is high (disabled) on the ADT7310. That is, can I connect many ADT7310 sensors to the same input pin on my microcontroller or do I need some logic (buffer) ta handle this?