The data sheet for the ADRF5731 shows a supply current of 117,-117 microamperes. Is this correct ??? as it seems to be "low". Any input is appreciated.
We have already updated the datasheet for ADRF5720.
The datasheets for the remaining products (ADRF5730/21/31) are being currently under revision and are going through the internal approval process at the moment. These should be completed shortly…
It appears that this may have been indirectly answered in a previous post when talking about the ORx port:
My question is with Rx2 confirming…
What are the differences between glitch free, safe state and no protection transitions in digital step attenuators (DSA)?
Digital Step Attenuators usually have an array of attenuator stages internally and turning on or off the different stages…
For a Digital Step Attenuator (DSA), what is step error and state error?
Step error: The step error is the error in stepping between consecutive states.