Original Question: Regarding ADP5301 by dinesh_10097
I am Dinesh ,I'm using ADP5301 Regulator to generate 3.3V from battery supply and I would like to verify the design from Analog devices Application engineer.
All things are as expected from…
Thank you for sharing the information.
Since the high efficiency at ultra light load is very important to me, and ADP5301's performance seems better than ADP5304.
So, I'll buy ADP5301 for my system. Thank you!
The ADP5301 cannot transit from hysteresis mode to PWM mode during MODE pin being low.
I would like to use ADP5301ACBZ-2-R7, Option 0 without output discharge. But I would like to use a 1.1v or 1.0v output. This output value is only available in Option 1.
Is it still possible to make a resistor divider outside the VID pad in order…
if you want to get ta 3.3V output from the battery, you can use a ultra-low power LDO ADP165/165 or switching regulator ADP5301 connecting to SYS.
The datasheet for the ADP5301 has 3 part numbers listed in the ordering guide that detail what the factory options are for voltage and quick output discharge but do not state the soft start option. Is the assumption that all parts are at the default soft…
My evaluation board is configured for PWM (Sync/Mode pin is at Vin). The output is set to 3.0 volts and the input is 5.0 volts. With no load, the current draw is 5.75 mA. If I configure the regulator for hysteretic mode, the current draw is per the datasheet…
REG_OUT is regulated output that can be selected by VID but SYS is a floating output that can be charged up to TERM threshold. if you want to use this output, it's better to connect ADP5300 or ADP5301 at SYS pin controlled by ADP5091 PGOOD.…
可穿戴医疗保健设备可定义为能够自主监控或支持特定医疗功 能的无创式系统。(可穿戴的国家标准正在制定中) “可穿戴”一词说明该设备要么是直接佩戴在人体上的设备， 要么是一件衣服，并且具有适当的设计，支持其用作可穿戴配 件。广义上讲，这要求该设备具有极小的尺寸和重量，功能自 主，自备电源，易于使用，而且穿戴舒适。