Revision D of the ADIS16375 datasheet has captured these improvements and is now online.
I was able to try this on an ADIS16480 and it worked as expected. The ADIS16480 has a similar function to the ADIS16375. Unfortunately, I am out of the office and don't have an ADIS16375 with me. Any updates on your end?
We are using both ADIS16485 and ADIS16375.
We have succeeded to activate the ADIS16485 IMU with our Controller.
We have made some dynamic tests on the IMU calibration jig and got reasonable results.
We could not make the ADIS16375 unit work…
The ADISUSB evaluation system provides some PC-based evaluation tool support for the ADIS16375. Click on the following Wiki Guide for instructions on evaluating the ADIS16375 on the ADISUSB:
Thank you for the support very quickly. I refer to the demo project for
ADIS16227, but I have adjusted accordingly according to the datasheet of
ADIS16375. thank you for sharing my project ADIS16485 with teensy on
arduino. I will try that priject to…
The ADIS16375/PCBZ is now obsolete. New users can replace this function by ordering the ADIS16375BMLZ and the ADIS16IMU1/PCBZ. Click on the following link to access a link for online purchase of these two models:
6g is within the measurement range of the ADIS16375 and in general, is pretty benign for PCB construction materials. You might want to consult with AN-1295. This was the design guide for mechanical interfaces with the ADIS1648x products, but it applies…
Janne was kind enough to share that his system had an issue with a regulator that was passing +5V through to the ADIS16375 and that once he fixed the regulator, restoring the ADIS16375's supply voltage to +3.3V, normal operation was restored. Many…