I'm looking forward to design an ultrasound imaging system using phased array transducers(preferably 2D). I need to figure out the best AFE design for my need. I assume that AD9670 + HSC-ADC-EVALCZ Data Capture Board
would be suitable…
I was hoping to use the AD9670 for my honors project, how can I get access to the full datasheet?
On the AD9670 data sheet, page 6, the current consumption for some of the rails is specified as:
IAVDD1, TGC mode, LO band mode: 148/187/223/291 mA
Under what conditions is the current consumption 148mA, and what conditions would cause it to be 291mA…
To Whom it May Concern,
Question regarding the low pass filter frequency. On preliminary documentation, the low pass filter was supposed to go as high as 26MHz at 80MHz clock. The SP0 version datasheet now shows a number of frequencies “Out of tunable…
What is the reason for the having the demodulator after the ADC in the functional block diagram?
I want to buy AD9670 Evaluation Board (AD9670EBZ), so I'd like to know have you got a reference design for it,
same as for AD9671 Evaluation Board, but using Serial LVDS to connect with FPGA ?
I have a problem to correct configure tha AD9670.
If I configure the FCO in "continous mode", the FCO and the DCO signal go out correctly but the data(DOUT) are always zero. Istead if I configure FCO "only during burst" the DCO signal go out…
I would like to use AD9670.
So, I have some questions.
#1 What is the purpose of the EXTERNAL SYNC pin of the capture board (HSC_ADC_EVALZC)?
#2 Is data output correct as it is output from 16 data output pins of FPGA?
#3 When sending the signal…
I would like to know the good configuration to cancel the DC offset after the ADC (in numeric part).
I try this with no effect. I always have an offset between (-20 and -10 (decimal)).
I sent 0x0000 (reset) and just next tap 0x0001 (set on)…