Is it possible to do Chemical analysis with Arduino, Linduino One in combination with EmStat Pico?
Using an Arduino board in conjunction with the PalmSens EmStat Pico evaluation board or module is easy due to the serial communication port via UART. The evaluation board inlcudes also a bluetooth module.
The example code provided by PalmSens is made for the Arduino MKR. The difference to the Arduino UNO or Linduino One is that the Arduino MKR board has a physical second UART interface available. The baud rate of the PlamSens Module is 230400 baud, this data rate can't be achieved by using software TX/RX. It is not foreseen that the customer is changing the data rate at the PlamSens module.
MethodScript Code Examples can be found here: