In datesheet，only Sensitivity of LSB/°/sec is given.(200 for 290 and 80 for 450)
Can you give an example of calculating the measurement error by measuring a certain value, such as 10°/sec?
Thanks a lot.
The datasheet also specifies the offset error, sensitivity tolerance and change with temperature, noise, non-linearity and cross axis sensitivity as sources of error.
All these parameters add to the…