I am working using ADE9153A. If I use Rshunt of 1mOhm then everything looks fine but when I replace it with 0.5Ohm then after calibrating the value 10 times the actual. to change the Rshunt value?
I tried recalculating the AIHEADROOM value of 8,83884 but the result is still inaccurate.
Thank for help.
Did you mean 0.5 mOhm? 0.5 Ohms is very large for a current sense resistor, it means that the output signal of the shunt will be 500 times larger and that could certainly pose a lot of issues.
If you did mean 0.5 mOhm, can you please explain in some more detail what the issue is? Have you made sure to modify your TARGET_AICC to compensate for the size difference in the shunt, this is important in the calibration process. If you run MS_ACAL_AICC, does the value you are getting back seem reasonable, I would expect it to be twice the value as with the 1 mOhm shunt.