QThe customer wants to use ADG936 because of its low insertion loss. But in the
datasheet, the figures about insertion loss vs frequency are all from
30KHz~10G. And the customer wants to know the information between DC to 50MHz.
Could we provide it to the customer? If yes, please send it to us. Also pls
refer to the following email.
AThe reason the frequency sweep is from 30kHz is that this is the lowest
frequency our test instrumentation will go to (R&S VNA). From what we have
seen the IL is flat at lower frequencies and it will be in at a 0.4dB typ
level. See the specifications table for more details.
1 person found this helpful