the thermal drift. The datasheet specifies an offset drift of +/- 2ppm/°C.
Could you please let me what unit the ppm are referring to ? Is it full scale
input voltage (2V)?
For the offset it is full scale range, or 2Vp-p.
Offset drift = (drift coefficient %FSR/⁰C) * DT = ±2PPM %FSR * DT = ±2/1E6 *2
*DT = ±4uV* DT, or ±4uV /⁰C
We did not specify a gain drift coefficient on this device, but we do
guarantee all channels gain error and channel to channel matching over