- We are getting erroneous conductivity readings with ADuCM3029_demo_cn0428_cn0429.
- This is with the latest hex file downloaded 4/1/2021.
- We are using standard solution KCL-700, a 700 microsiemens solution.
- We are using the recommended probes - Cole-Parmer 100-series Replacement Conductivity/Temperature Probe, K=1, 59200-06 probes
- DIP switches are set to Z mode.
- Mode is changed to conductivity.
- We are getting a reading readings like 3e-07 S/cm (screenshot).
- We should be getting readings like 7e-04 S/cm.
Thanks for your help.