Today I tried to assemble the second unit of our gas analyzer prototype. I used exactly the same hardware and to my surprise it didn't work, I couldn't even turn it on. After a review and finding nothing missing, disconnect the sensors with the CN0429 boards, connect the power and the system start normally. If I reconnected the sensors, the system would shut down as if I suffered some kind of electrical problem (short circuit).
Looking carefully at the EVAL-M355-ARDZ-INT board in prototype 1 and prototype 2, I realize that in the second all the connectors were welded upside down (see photo). The board that works correctly says to be assembled in the USA while the other does not indicate where it was assembled. The lousy production quality control system that Analog Device is having is inconceivable.
I urgently need a way to return the defective boards EVAL-M355-ARDZ-INT and EVAL-CN0429. I really feel disappointed by the situation
Split from a previous conversation because its a new question with a different title
[edited by: Brandon at 3:16 PM (GMT 0) on 23 Sep 2019]