We have set up EVAL-CN0359 and have the following problem. We are measuring the voltage from J5 pins 2 or 3. These feed AD8253. We see the expected distorted square wave on the scope but it is floating up and down between the rails. Page 18 of the datasheet for AD8253 possibly implies that the inputs are not stabilized by a bias current path to ground in the CN0359 layout. See also fig. 6 of your article by Kitchin concerning "Common Problems When Designing Amplifier Circuits". Any advice here? Thanks--Fritz
I would like to check a couple of things on the board.
1. Reference voltage on pin 9 of U15 w.r.t. ground. This should be ~1.65V
2. Short pins 2 and 3 of J5 to ground and measure output of U15. This should give you ~1.65V at the output of the amp.
3. What is the coupling and impedance of the probe used? This should be DC coupled and set to high impedance to avoid loading of the probe.
4. Can we try increasing the frequency used for the measurement? We are currently at 1kHz and the specified range of your cell is from 100Hz to 10kHz.
1- pin 9 shows 1.64
2- The output of U15 with 2&3 at gnd shows 1.828V
3-Our probe is 10M 11pf
4-We did the above at 0.3V and 1kHz. At 5.1kHz it is roughly the same, the U15 output showing 1.838V
We did all of this on a new board. Looking at the output of U15 at 5.1kHz, 0.3V and with the IST sensor, we still see the gradual drift to the rails.
Any further action on this? We are experiencing plating on our probes, very likely due to this phenomenon. Are you equipped to set up this circuit in your lab and duplicate our results?
Apologies on the delay.
Based on the data that you sent. The system looks to be working fine. I am thinking if the probe that you have could be not functioning the way it is. This could be the overvoltage condition on the earlier part and the added plating on the probes. What kind of plating do you have on the probe?
Do you happen to have an extra probe to try this out? We have the board but do not have your exact same probe.
Glenn-we had replaced the failed probe some time ago and are running 2 or 3 of you boards with fresh probes, all exhibiting the same behaviour. We also have run the CN0359 with a commercial TOPAC probe and it similarly shows this problem.
I do not quite understand how the system can work with the common mode drifting between the rails and indeed when it does this an error condition is raised by your software.
Where are you located? We can send you a few test probes.
Meanwhile have you tested this with a probe of your own and watched the voltages with a o'scope? If you see the problem you can continue analyzing it. If not we can buy the same probe and do some tests with it here.
Also is it possible to communicate via telephone? We lose a whole day between each exchange otherwise and we can quickly try the things that you suggest over the phone.