I want to ask about the impedance measurement range of aducm355. I saw in the introduction at the beginning of the document that it is <1-10MΩ,
but in the parameter table below(page 7), it is only 10000Ω, which is 10KΩ.
Hi, The actual range is from <1Ohm up to 10Mohm however the accuracy specification applies to the range of 0.4Ohm to 10kOhm. What range are you interested in?